The EC help provides an easy access to the information related to the use of Keysight EC Network Analyzer. Pressing Help key on the front panel. EB/EB, please see the Programming Manual. • Installation and Quick Start Guide (Part Number: Ex1, attached to optional ABA). This manual . The EC complies with INSTALLATION CATEGORY II as well as The Service Manual is a guide to servicing the EC ENA Series Network Analyzer .
|Published (Last):||9 January 2014|
|PDF File Size:||7.67 Mb|
|ePub File Size:||4.52 Mb|
|Price:||Free* [*Free Regsitration Required]|
The following screen appears during backup. Click Load to start the calibration measurement.
Atec Agilent-EC User Manual | 60 pages
Select the C drive to restore, then click Next button. After a while, the EC restart automatically.
E Figure 4 Item 4: The error correction manua will also be automatically enabled. The volume label for C: There are two procedures in accordance with the HDD revision. All rights are reserved.
If you press F11 twice or more and a screen which is not mentioned in this manual appears, reboot the instrument and restart the process from the first.
E Removal Procedure Step 1. When the recovery is finished, push the enter select OK. CN or above Step 1. Select the C drive, then click Next button. The User back up function is not available. Click Short to manula the calibration measurement. Inset the key in the lock and turn anti-clockwise.
When the screen as shown in the figure below appears, press F11 of the keyboard. CN or below This section describes how to create the user backup image. Dangerous voltage levels, capable of causing death, are present in this instrument. Types of system recoveries The following 2 types maunal system msnual are available.
The EC restarts automatically. Recovery function is performed. Select the C drive and D drive. When the recovery is finished, EC restarts. DO NOT Service or Adjust the Instrument Alone – Do not attempt internal service or adjustment unless another person, capable of rendering first aid and resuscitation, ,anual present.
You can also contact one of the following centers and ask for a test and measurement sales representative. Close all other applications you currently open. If the name label of the calibration kit has been changed, the label is displayed as the soft key. The following screen appears when the user backup image was created.
Component replacement and internal adjustments must be made by qualified maintenance personnel. CN or below Step 1. The System Properties dialog box appears.
Figure 3 Item 3: Turns on the EC. Click OK to proceed. Call Center tel tel 81 fax 81 Korea: Factory recovery Returns the contents of the C drive to the factory state. Network setting GPIB setting Printer setting The driver for the supported printer installed after purchase is deleted.
If you use a calibration kit other than a predefined one, you need to define it. Do not replace components with the power cable connected.
Repeat the above procedure for port y. A dialog box appears to allow you to select a program to be loaded. Click Cal Kit, then select the calibration kit.
Agilent Technologies assumes no liability for the customer’s failure to comply with these precautions. Select a test port and corresponding S parameter on which 1-port calibration will be performed. Remove the hard disk. Amnual message dialog box appears after the completion of the Recovery Image Setup procedure. Turn the key to the left until it is locked.
Out-position of push-button switch. To avoid injuries, always disconnect the power and discharge circuits before touching them.