ii Heracell Service Manual – Free download as PDF File .pdf), Text File . txt) or read online for free. incubator heracell SM. HERAcell® CO2 Incubator Operating Instructions. Version: / – 3 -. Contents. 1. Information regarding Safe Operation. Manuals and User Guides for Thermo Scientific Heracell i. We have 1 Thermo Scientific Heracell i manual available for free PDF download: Operating.
|Published (Last):||26 October 2006|
|PDF File Size:||2.87 Mb|
|ePub File Size:||5.12 Mb|
|Price:||Free* [*Free Regsitration Required]|
Chapter 2 Delivery Of The Device O gas cylinder 7. The system returns to the Configuration menu. Room Ventilation When installing the device, make sure that the installation and supply connections are freely accessible. Water pump Water pump Figure.
Examples Of Data Logger Codes 1. Set value CO 9. EXE to open this file.
Page 50 A pressure compensation opening with insert in the rear panel of the device allows a compensation between the pressures in the work space and in the operating room. Temperature Calibration Procedure  in the rear panel of the device. CO gas cylinder 8. Make sure to prevent Contamination when operation is interrupted. Gas Connection Set the gas supply to a range between 0. To ensure the accessibility for care and maintenance works, keep larger side and rear distances.
Heat-up Phase Of The Control Loop Sensors When the device has been switched on, the control loop sensors pass through a heat-up phase of varying duration during the start process: The center of the work space is the reference location for the comparison measurement.
Interruption Of Operation Their measured values 240k compared to the selected set values. Replacing the door seal Work space components Work space components Figure.
I have 2 of 4 Heracell ‘s purchased in whose fan motors died within 1 month of each other. Page 36 3 additional devices independent of device type Page When the remaining run time has elapsed to 0 hours, the device has reached the originally set working temperature again e.
Until the device is shut down, the work space must be continuously ventilated.
When the shelves are being drawn out, ultimately destroying the samples. Decontamination routine phases Figure.
Thermo Scientific Heracell 240i Manuals
Notice The contra-con decontamination routine must be performed every 3 months. Connecting The Usb Interface optional Install hose  to the sleeve of the gas humidification and then to the sleeve  of the device-integral oxygen or nitrogen supply line. Decontamination Procedures The following disinfection procedures are compatible with the device: Turn the PC on.
To directly access the Bottle turning device submenu, touch the Bottle turning device icon in the main menu. Operational Safety Rules Standards And Directives Press the power switch; the heracfll illumination goes off. Note Duration of the data transmission: Examples Of Data Logger Codes To indicate the start of this routine, the icon overtemperature is displayed.
To allow a rapid disconnection of power in case of an emergency, make sure that power outlets remain freely accessible at all times!
Thermo Scientific Heracell 240i Operating Instructions Manual
Do not spray disinfectant onto the CO sensor in the measuring cell baseplate or onto the O sensor. Wipe the interior surfaces ans parts with a diluted quaternary ammonium disinfectant. The connection to the power supply system is established by connecting manul cable with plug for IEC hercaell  to the socket .
Intended Purpose Of The Device Page 80 – Operating panel and operating structure Page 81 Page 82 Page 83 – Structure of the operating levels Page 84 – Heat-up phase of the control loop sensor Information on these tests is available from Thermo Scientific upon request.
Instructions of use, instructions for Thermo HERAcell-CO2-Incubator i i manual en
Page Chapter 14 Device log For a detailed overview of causes for errors and their correction, please refer to the end of this chapter! Note Make sure that power sockets remain freely accessible at all times! Connecting The Rs Interface Power Supply Connection Chapter manua, Data Communication To toggle between two options :